SILVER SPRING, Md., Sept. 22 /PRNewswire-USNewswire/ -- The U.S. Food and Drug Administration today cautioned users of personal emergency response buttons worn around the neck of a potential choking hazard associated with this product.
The FDA is aware of at least six reports between 1998 and 2009 of serious injury or death, including three deaths in the United States and one in Canada, from choking after the cord on the Philips Lifeline Personal Help Button became entangled on other objects worn around the neck.
There are more than 750,000 users of these devices in the United States and Canada. By pushing the "help" button on the device when in distress, users can call for emergency assistance to their home. According to Philips Lifeline, the device is used primarily by seniors living independently, who feel they are at risk for falls or other medical emergencies.
The Lifeline pendant button is intentionally designed to not break away when tugged, which prevents the button from accidentally falling off. However, because it does not break away, there is a risk of choking, including the possibility of serious injury or death. Risks are greater for those with mobility limitations or for those who use wheelchairs, walkers, beds with guard rails, or other objects that could entangle with a neck cord.
Philips Lifeline is currently sending letters to its 750,000 customers and has changed the labeling of this product to include a warning against the potential choking hazard.
The FDA recommends that users consult their health care providers to determine which style of emergency button, including those that are worn on the wrist, is most beneficial for them.
These widely used devices provide critical and immediate access to emergency care for those at risk of falls or who may be more likely to need outside assistance. While the number of adverse events reported is small compared to the number of people who use this device, the severity of these events is of concern. It remains important that users, along with their health care providers, assess the options provided by each style of button, and choose the option that best fits their condition.
Health care professionals and users may report serious adverse events or product quality problems with the use of this product to the FDA's MedWatch Adverse Event Reporting program either online, by regular mail, fax or phone.
Online: www.fda.gov/MedWatch/report.htm Regular Mail: use postage-paid FDA form 3500 available at: www.fda.gov/MedWatch/getforms.htm and mail to MedWatch, 5600 Fishers Lane, Rockville, MD, 20852-9787 Fax: (800) FDA-0178 Phone: (800) FDA-1088 Media Inquiries: Peper Long, 301-796-4671, email@example.com Consumer Inquiries: 1-888-INFO-FDA
|SOURCE U.S. Food and Drug Administration|
Copyright©2009 PR Newswire.
All rights reserved