In all, the device looks like a miniature sardine can with a porthole.
But what the team has packed inside makes it a major advance among brain-machine interfaces, said lead author David Borton, a former Brown graduate student and postdoctoral research associate who is now at Ecole Polytechnique Federale Lausanne in Switzerland.
"What makes the achievement discussed in this paper unique is how it integrated many individual innovations into a complete system with potential for neuroscientific gain greater than the sum of its parts," Borton said. "Most importantly, we show the first fully implanted microsystem operated wirelessly for more than 12 months in large animal models a milestone for potential [human] clinical translation."
The device transmits data at 24 Mbps via 3.2 and 3.8 Ghz microwave frequencies to an external receiver. After a two-hour charge, delivered wirelessly through the scalp via induction, it can operate for more than six hours.
"The device uses less than 100 milliwatts of power, a key figure of merit," Nurmikko said.
Co-author Ming Yin, a Brown postdoctoral scholar and electrical engineer, said one of the major challenges that the team overcame in building the device was optimizing its performance given the requirements that the implant device be small, low-power and leak-proof, potentially for decades.
"We tried to make the best tradeoff between the critical specifications of the device, such as power consumption, noise performance, wireless bandwidth an
|Contact: David Orenstein|